1. Assist in observing the state of the clearing sample;
2. Accurately determine the degree of sample clearing and the size of the sample;
3. Assist in observing and locating impurities and air bubbles in the sample and assist in sample trimming and other operations.
1. Open the package, check the condition of the instrument, and the number of accessories. If there is any problem, contact us immediately.
2. Put the Backlight Plat on flat table to avoid falling and damaging it.
3. Connect the BackLight Plate, USB cable, power adapter, and power supply. When the blue indicator at the interface between the BackLight Plate and USB cable is on, it indicates that the BackLight Plate is powered on. (If the power indicator is not on, replace the USB cable or power adapter)
4. After power on, wait 3 seconds, touch the switch to turn on.
5. Touch the power button to adjust the brightness, the instrument has three brightness adjustable.
Nuohai BackLight Plate For Transparent Tissue .pdf